Updates on the new standard for enteral feeding connectors
(ISO Standard 80369-3)
What’s happening to enteral feeding connectors?
Cook Medical and other medical device manufacturers around the world are transitioning to new ISO standards for small-bore connectors in healthcare applications, starting with the standard for enteral feeding connectors. These new standards have been created in order to enhance patient safety and reduce harmful misconnections between medical devices.
How do misconnections happen?
A medical device misconnection can cause medication, oxygen, or fluid to be delivered to the wrong place. Misconnections can happen for many reasons, but one key reason is because connectors in one type of medical device are physically compatible with another type of medical device or accessory. The new ISO standard 80369-3 will require feeding tube connectors to be usable only with enteral feeding devices, in order to reduce the chance of misconnections.
How is Cook involved?
Cook is collaborating with other medical device manufacturers to adopt the ENFit® connection, which requires a twisting motion in order to connect, on its feeding tubes. The two ends of the ENFit connection are designed to fit only enteral devices. The ENFit connection meets the new
ISO design standard.
The path to safer connections will happen in three key steps:
Step 1: Meeting the new standard
Cook Medical is committed to enhancing patient safety by reducing potentially harmful misconnections between medical devices. That’s why we are fully supporting the initiative to transition to the ENFit connection. Plans are underway to incorporate the ENFit connection on all Cook enteral feeding devices. (See Step 3 below.)
Step 2: Ensuring ENFit compatibility during the transition
The entire enteral feeding industry is in the process of transitioning to the permanent ENFit connection. This will affect syringes, administration sets, and enteral feeding tubes. The ENFit Transition Connector (shown below) allows our current feeding tubes to be compatible with new ENFit syringes and administration sets already on the market. Please contact your local Cook sales representative if you need more information on the ENFit Transition Connectors and how to access them.
ENFit Transition Connector
Step 3: Completing the transition to the permanent ENFit connection
The permanent ENFit connection will be completely integrated into all Cook feeding tubes in the near future. Customers can transition to ENFit on their own timeframe. To determine where Cook products are in the transition, please contact your local Cook sales representative.
How to find out more
Cook is a member of GEDSA, the Global Enteral Device Supplier Association, which was formed to help introduce changes in international standards in medical device tubing. Review these GEDSA resources and share this important information with your institution.
This video, produced by GEDSA, provides a quick, informative overview of the expected enteral feeding connector changes:
GEDSA’s “Stay Connected” communication program will provide ongoing and timely information on the coming changes. On the “Stay Connected” website you can find Frequently Asked Questions, access information resources, and sign up for alerts. Click on the image to read more.
ENFit is a federally registered trademark of GEDSA in multiple jurisdictions throughout the world. Used with permission.